Instrument Staining and Pitting of Surgical Instruments
A brown/orange stain or a blue-black stain can occur from plating during the cleaning or autoclaving process. Through electrolysis when dissimilar metals touch while being autoclaved, ultrasonically cleaned, or sometimes even stored together, plated stains actually bond the stain material to the instrument metal. They do not often change the metal material except for the discoloration. These stains are very difficult to remove and should be sent to a surgical instrument service facility for refinishing.
Black stains are usually due to an acid reaction. An acidic detergent deposit left on the instrument during autoclaving might cause a black stain. Always use neutral pH detergents and distilled deionized water in your autoclave process, and be sure to completely rinse instruments before autoclaving.
Multi-colored stains or chromium oxide stains result from excessive heat. These rainbow colour stains indicate the instrument may have lost some of its original hardness after being heated. Cutting edges lose their sharpness quickly when hardness is reduced. Flash flame decontamination (an instrument is decontaminated by inserting it into a flame for a few seconds) changes the molecular structure of most material adversely, shortening the useful life of instruments.
Avoid the problem of pitting by using only pH neutral (7.0) detergents and making sure all instruments placed in the autoclave are thoroughly rinsed before being put in the autoclave.
Pitting can also occur if dissimilar metals come in contact with each other in an ultrasonic cleaner or autoclave. Electrolysis from dissimilar metals touching in a solution (the steam in the autoclave acts as a conductive solution that allows electrolysis) transfers metal molecules from one instrument to the other, leaving pits in one instrument.
Avoid having any instruments touching during autoclaving, cleaning or storage to eliminate pitting.